(Ref. 128118-001 REV. E)
0.8mm (IEC 336) standard
0.5mm (IEC 336) optional
Ultem: 3.00mm ±0.15mm
Oil: 7.5mm ±0.25mm
Glass: 1.7mm ±0.2mm
Fan: Standard. The beam angular coverage will be 80° with the beam plane perpendicular to the X-Ray tube axis and 20° wide (with a 2° tolerance)
Cone: Optional. 20° cone beam (with a 2° tolerance)
Power factor corrected input 0.98, 90-264Vac, 47-63Hz, 2A maximum
X-Ray Tube Voltage:
Nominal X-Ray tube voltage is adjustable 40kV (±20kV) to 80kV (±40kV)
X-Ray Tube Current:
150uA to 2.00mA over specified tube voltage range (100W max.)
X-Ray Tube Power:
100W, maximum continuous
Line: ±0.05% of maximum output voltage over a ±10% change of nominal input line voltage
Load: ±0.1% of maximum rated voltage for 150uA to 2.00mA load change
Voltage measured across the X-Ray tube is within ±2% of the programmed value
Standard: Ramp time shall be <500ms from 10% to 90% of maximum rated output voltage
0.5% peak to peak of maximum voltage for frequencies ≤1kHz
Emission Current Parameters
Line: ±0.05% of rated output current over a ±10% change of nominal input line voltage
Load: ±0.1% of rated output current for a change from 50% to 100% of rated output voltage
Current measured through the X-Ray tube is within ±2% of the programmed value
Standard: Ramp time shall be <500ms from 10% to 90% of maximum rated current
4 arcs in 10 seconds with a 100ms quench/100ms re-ramp = Shutdown
Internal AC filament drive with closed loop filament emission control
Ground referenced 0 to 9Vdc for all monitoring signals.
Relay contacts and open collector signals for other signals. See analog interface connector pin out table.
The RS-232 interface allows for programming of kV, mA output and X-Ray enable. Provides monitoring for kV, mA output and oil temperature. Tolerance 3%. (with an additional 5μA offset at ≤10% mA programming)
A demo GUI is available for engineering evaluations
0°C to +40°C
-40°C to +70°C
10% to 95% relative humidity, non-condensing
X-Ray Tank: Customer provided 250 cfm external cooling fan as required to maintain oil temperature below 55°C.
Controller: Forced air via internal fan.
Input Line Connector:
3-pin Phoenix Contact 1829167. Mating connector provided with unit
Analog Interface Connector:
15 pin Male D connector provided with unit
Digital Interface Connector:
9 pin female D connector, female
8-32 ground stud provided on chassis
X-Ray Tank: 11.3˝L X 9.625˝W X 4.93˝H (287.02mm x 244.4mm x 125.2mm
Controller: 8.5˝L X 6.70˝W X 2.21˝H (215.9mm x 170.2mm x 56.1mm)
X-Ray Tank: 36lbs (16.32kg)
Controller: 3.7lbs (1.68kg)
Can be mounted in any orientation.
Not to be greater than 0.5mR/hr at 5cm outside the external surface.
Compliant to EEC EMC Directive. Compliant to EEC Low Voltage Directive. UL/CUL recognized file E235530.
CB Cone Beam