Cargo screening, cancer therapy
0 to -15kV
Pulsed Cathode Current:
200 to 1500mA
Mean Cathode Current:
Current Pulse Width:
Up to 6μs (90% to 90%)
Current Pulse Frequency:
Up to 500Hz
0 to -6.5Vdc (wrt Cathode)
-60 to -135Vdc (wrt Cathode)
0 to 340Vdc (wrt Cathode)
Injector Current is regulated by programming
the Grid Pulse Voltage
Ethernet (RJ45 jack)
Interlock (24V nominal, 9 pin D-sub male pin plug)
Pulse Input (BNC)
AC Power In (IEC320 C14 male pin receptacle)
HV Out (3 pin terminal block)
24V must be present to enable output high voltage cathode. It
is intended to connect to the system to disable the ICA in case
any system interlocks have not been satisfied.
10°C to +40°C
-40°C to +70°C
30% to 60% relative humidity, non-condensing
The ICA shall be designed in accordance with IEC 60601-1
(Edition 3) Requirements. Certification shall be by agreement.
The ICA shall be UL 94V-0 and RoHS compliant.
||CONDITION Illuminated When...
||Input trigger is present.
||AC supply voltage is present.
||Interlock in signal is present.
||Arc fault signal is active.
|Cathode Current Fault
||Cathode current fault signal is active.
|Cathode Voltage On
||Cathode high voltage is present.